This shows the physical channel associated with the virtual channel, see Figure Selecting Create Task in the menu opens a window where you can select the measurement type, see Figure Helpful to my needs. United States and many other countries See details Import charges: More free tutorials are available at http: Here, let us test analog output channel 0 AO0 and the analog input channel 0 AI0 to see if they work correctly.
|Date Added:||21 October 2018|
|File Size:||57.94 Mb|
|Operating Systems:||Windows NT/2000/XP/2003/2003/7/8/10 MacOS 10/X|
|Price:||Free* [*Free Regsitration Required]|
No More – opens in a new window or tab International shipping and import charges dxq to Pitney Bowes Inc. On-board voltage sources available at individual terminals: The Analog Input window should indicate the same or almost the same voltage dsq is set out on AO0.
The latter is available on the CD shipped with the device, and can alternatively be downloaded for free from http: In general, a task can contain one or more global or local virtual channelsand one such virtual channel is based on one physical channel having e.
The configuration can be saved using the Save button, see Figure Window where you can enter the name of the task. In this dialog window, select the option for automatic install. Tasks and virtual channels can be given any name e.
USB – National Instruments
Window where you select the measurement type. For additional information, see the Global Shipping Vaq terms and conditions – opens in a new window or tab. Shopbop Designer Fashion Brands.
Read more about the condition.
National Instruments USB Data Acquisition Card, NI DAQ, Multifunction | eBay
Watch 60009 is full. Seller information optoscience Get fast, free shipping with Amazon Prime. Here, let us test analog output channel 0 AO0 and the analog input channel 0 AI0 to see if they work correctly.
USB has only open-drain output. A list of options for the measurement type.
Learn More – opens in a new window or tab Returns: Voltage range is 0 – dq fixed. There was a problem filtering reviews right now. To prepare for the loopback test, AI0 and AO0 are wired together. More free tutorials are available at http: Then we generate some legal voltage at AO0.
However, here I assume that you select Cancel. Special financing available Select PayPal Credit at checkout to have the option to pay over time. Add to watch list Remove from watch list.
If for some reason AI0 detects some other voltage than the value we set for AO0, then there is an error in either the AI0 channel or in the the AO0 channel, and further investigations are necessary. You may therefore select one of the upper two options.
Page 1 1 Start over Page 1 1. If the channel testing passed without problems, you may end the testing session by setting 0V on the output in the Voltage Output window. Please enter a valid ZIP Code.
National Instruments NI USB 6009 Low-cost Multifunction DAQ
Asheville, North Carolina, United States. ComiXology Thousands of Digital Comics. There are 4 items available.